Please use this identifier to cite or link to this item:
|Other Titles: ||Detection of NADH with nanosheet of Cobalt(II,III) oxide based modified in flow injection analysis system|
|Authors: ||陳映慈;Chen, Ying-Cih|
|Keywords: ||四氧化三鈷;流注分析系統;NADH;Cobalt(II,III) oxide;Flow injection analysis system(FIA)|
|Issue Date: ||2012-06-21 06:37:53 (UTC+8)|
本研究中進一步探討流注分析系統(FIA)以評估偵測NADH時的最佳化條件：70% Co3O4修飾電極，偵測環境為0.2M pH 6.00 磷酸鹽類緩衝溶液，偵測電位為100 mV (vs. Ag/AgCl)，載體流速0.25 mL/min，樣品迴路體為50μL。偵測NADH之分析特性分別為：線性範圍為10-100μM (R=0.99878)，電流密度為0.35899 nA/μM，偵測極限(S/N=3)為1μM，連續重覆二十次偵測NADH，所得到相對標準偏差(RSD)為1.4%，反應時間(t90%)為12.26s；在0.1V的電位下，大部分的易氧化物質如dopamine、Urea、 (±)-epinephrine、serotonin、histamine、4-acetaminophen等，均不會有明顯干擾本系統之偵測，僅有抗壞血酸(AA)造成干擾，因此在偵測前先加入抗壞血酸氧化酶進行前處理，加入3U的抗壞血酸氧化酶處理五分鐘之後進行偵測，可避免AA之干擾。
NAD+/NADH is an important cofactors for a class of apozymes in several dehydrogenase, which involves in more than 300 biological reactions. Therefore, a NADH based sensor possesses a great potentiality in development of other biosensors. However, the prior studies indicate that direct oxidation of NADH requires a high overvoltage over 0.7 V, which would be suffered from most of all common biological antioxidants. Besides, a significant surface fouling effect due to the strong adsorbability of the oxidized intermediate also causes a bad reproducibility and low sensitivity. In this work, we develop a simple, sensitive and reliable NADH sensor based upon the Cobalt (II, III) oxide modified electrode. Owning to the redox property of passive layer of the cobalt oxide, the oxidation potential of the NADH can be shifted from 700 mV to 100 mV.
In order to develop a high sensitive scheme for NADH determination, this sesor is held by using a high efficient flow injection analysis system. After optimization, 70% cobalt (II, III) oxide modified carbon ink electrode with a constant operating potential of 100 mV (vs. Ag/AgCl) in 0.2 M phosphate buffer, pH 6.00, and flow rate of 0.25 mL/min were chosen as the optimal conditions. A suitable linear range 10 to 100 μM (R=0.999) with a sensitivity of 0.36 nA/μM is achieved. The detection limit base on 21 successive blank injections is 4.25 μM (S/N=3).the relative standard deviation (RSD) for 20 successive measurements of 75 μM NADH is only 1.4%, which indicates the high stability of this method. Besides, most antioxidants such as dopamine, uric acid, (±)-epinephrine, serotonin, histamine, and 4-acetaminophen do not affect the NADH determination. However, ascorbic acid (AA) causes a significant interference to this scheme. Here, an ascorbate oxidase (EC 18.104.22.168) was utilized to oxidize the ascorbic acid before sample injected into the flow injection analysis system, and the interference from ascorbic acid was eliminated, successfully.
|Appears in Collections:||[化學學系暨研究所] 學位論文|
Files in This Item:
All items in 機構典藏 are protected by copyright, with all rights reserved.